2 edition of Calibrating two 6-port reflectometers with only one impedance standard found in the catalog.
Calibrating two 6-port reflectometers with only one impedance standard
Cletus A Hoer
by Dept. of Commerce, National Bureau of Standards, National Engineering Laboratory, Center for Electronics and Electrical Engineering, Electromagnetic Technology Division, for sale by the Supt. of Docs., U.S. Govt. Print. Off. in Boulder, Colo, Washington, D.C
Written in English
|Statement||Cletus A. Hoer|
|Series||NBS technical note ; 1004, NBS technical note -- 1004|
|Contributions||National Engineering Laboratory (U.S.). Electromagnetic Technology Division|
|The Physical Object|
|Pagination||vi, 39 p.:|
|Number of Pages||39|
The Michigan fur trade
Witness to our time
The sacred world of the Christian
winter of our discontent
Problem areas in the use of intra-oral auxiliaries in present public dental care programs in Ontario
family of Pascual Duarte
The only tradition
Get this from a library. Calibrating two 6-port reflectometers with only one impedance standard. [Cletus A Hoer; National Engineering Laboratory (U.S.). Electromagnetic Technology Division.]. This paper describes a technique for calibrating a pair of 6-port reflectometers for measuring the complex reflection coefficient of 1-port devices, or the scattering parameters of reciprocal 2.
Automated wideband test system, measurement uncertainty, and design of on-chip six-port reflectometers for 5G applications May DOI: /ARFTG Full text of "NBS calibration services users guide " See other formats.
Now, just two years after Naledi’s announcement to the world, a recent discovery w w s m a g a z i n e.c o. z a In Octoberscientists accidentally turned carbon dioxide (CO2) into. CXA Series analyzer.